Important Safety Information
If you use peritoneal dialysis solutions containing icodextrin (such as EXTRANEAL®, Icodial®) or take certain immunoglobulin therapies that contain maltose (such as Octagam® 5%, WinRho® SDF Liquid, or HepaGam B™) or other therapies that contain or metabolize to maltose, galactose or xylose (Orencia®, ADEPT® or BEXXAR®), your glucose meter results will be inaccurate if you use a GDH-PQQ enzyme-based meter system to measure the presence of glucose in your blood. These substances can interfere with the TRUEresult®, TRUE2go® and TRUEtest™ blood glucose monitoring products and can cause falsely elevated blood glucose results.
For further information on Nipro Diagnostics products, please call our technical support professionals at 1-800-803-6025.
If you have additional questions please read our Frequently Asked Questions.
For more information on EXTRANEAL (Icodextrin) and blood glucose testing, visit Baxter’s Glucose Safety Website at www.glucosesafety.com.
Healthcare professionals, please click here for further information.